Basics of maintaining and working with ion chromatographs.
The hardware is stopped in MagIC Net. Once the temperature has fallen down to room temperature and the pressure has fallen to 0 MPa, the column and guard column can be removed from the system. The leaflet for the column describes the optimum storage conditions for the column. A coupling (6.2744.040) is necessary to connect the capillaries.